Unlike some systems that require you to scroll through multiple numbers to code the meter manually, the code chip for the Accu-Chek Aviva Nano blood glucose. View and Download Accu-Chek Aviva Nano user manual online. Aviva Nano Blood Glucose Meter pdf manual download. Page 1. Page 2. Page 3. Page 4. Page 5. Page 6. Page 7. Page 8. Page 9. Page Page Page Page Page Page Page Page Page
|Published (Last):||11 August 2018|
|PDF File Size:||2.76 Mb|
|ePub File Size:||3.72 Mb|
|Price:||Free* [*Free Regsitration Required]|
The statutory provisions on rights in consumer goods sales in the country of purchase shall apply. Do not move the meter or the cable while the devices are communicating. Put the meter on a flat 5.
Maintenance and Troubleshooting Changing the Batteries Comments to this Manuals Your Name. The arrows on the screen indicate that the Accu-Chek Performa meter is in communication mode. Specifications Blood volume 0. Do not move the insulin pump or the cable during the communication between the insulin pump and the computer.
Coding the Meter 1. Before attempting to connect the Accu-Chek Aviva Nano meter for communication with the Accu-Chek Connect online diabetes management system, follow these recommendations:. You are now ready to use the Accu-Chek Connect online diabetes management system to communicate information from the Accu-Chek Voicemate voice unit.
If you do not have a user’s manual, please contact your local Roche representative. Page 59 Page 60 – Chapter 5: Page 51 50 Symptoms of High or Low Blood Glucose Being aware of the symptoms of high or low blood glucose can help you understand your test results and decide what to do if they seem unusual. If you still have problems, contact Accu-Chek Customer Care at The Accu-Chek Performa Insight diabetes manager, also referred to as a meter, communicates with the Accu-Chek Connect online diabetes management system through a USB cable that connects the meter to the computer.
The meter is in set-up The meter is ready for a mode, waiting for you drop of blood or control to change or confirm solution. The software is now ready to communicate information from the insulin pump. Viewing Test Results With the meter on or off, press and release to enter memory. Troubleshooting Checks Action 1. Check if the test strips are expired.
Doing so puts the Accu-Chek Aviva Nano meter in communication mode. Here is how to mark a test result and initiate a post-meal reminder: Test results are stored from the newest to the oldest. Page of 85 Go. Unless prompted, do not press any buttons on the device while in communication mode.
Window To transfer the results to a computer, connect the infrared cable according to the instructions. Page 72 The time and date settings may be incorrect. To use it correctly and dependably, you need to understand its operation, screen displays and all individual features.
Accu-Chek Spirit Click for instructions. Check if the code number on the meter If they do not match, insert the correct display matches the code number on code chip into the meter and test again. If this does test ,anual error has occurred. If you have any concerns, call Accu-Chek Customer Care at You are now ready to use the Accu-Chek Connect online diabetes management system to communicate information from the meter.
If the problem persists, call Accu-Chek Customer Care at If you need to clean it, follow these guidelines carefully to help you get the best avivx possible: Manuxl 15 Yellow Release Button 5. Viewing Test Results 7, 14, 30, and 90 day averages. The meter can also be put into communication mode by plugging the USB cable into the computer if Data Transfer is selected as the default communication mode. Verify that the code number printed on the test strip container matches the code number printed on the code chip currently inserted in the meter.
Maintenance And Troubleshooting Chapter 5: Accu-Chek Performa Connect Click for instructions. I need to order accu-cyek self test diary but I can’t order online can you let me know how I can get some. Page 65 Battery power is low.
ACCU-CHEK AVIVA NANO USER MANUAL Pdf Download.
Change the the system. Page 9 Golden End — Insert this end of accu-chke test strip into the meter. Accu-Chek Instant S Click for instructions. Manyal insulin pump automatically returns to the STOP mode 15 minutes after the last data transfer when the Data Transfer screen in selected.
Doing so puts the Accu-Chek Compact Plus meter in communication mode. Before attempting to connect the Accu-Chek Compact meter for communication with the Accu-Chek Connect online diabetes management system, follow these recommendations:.
You can review them at any time.
Devices Available Outside the United States
Wash and dry your hands. Make sure the time and date are correct and adjust, if necessary.
Perform a ensure the time and date in blood glucose or control test.